Swift's Burst Alert Telescope (BAT)For details of BAT Data Analysis, see our Data Analysis page: Swift Documentation section.
Instrument DescriptionThe Burst Alert Telescope (BAT) is a highly sensitive, large FOV instrument designed to provide critical GRB triggers and 4-arcmin positions. It is a coded aperture imaging instrument with a 1.4 steradian field-of-view (half coded). The energy range is 15-150 keV for imaging with a non-coded response up to 500 keV. Within several seconds of detecting a burst, the BAT will calculate an initial position, decide whether the burst merits a spacecraft slew and, if so, send the position to the spacecraft.
In order to study bursts with a variety of intensities, durations, and temporal structures, the BAT must have a large dynamic range and trigger capabilities. The BAT uses a two-dimensional coded aperture mask and a large area solid state detector array to detect weak bursts, and has a large FOV to detect a good fraction of bright bursts. Since the BAT coded aperture FOV always includes the XRT and UVOT fields-of-view, long duration gamma-ray emission from the burst can be studied simultaneously with the X-ray and UV/optical emission. The data from the BAT will also produce a sensitive hard X-ray all-sky survey over the course of Swift's two year mission. Below is a cut-away drawing of the BAT, and a table listing the BAT's parameters. Further information on the BAT is given by Barthelmy (2000).
BAT Table of Instrument Parameters
The BAT's 32,768 pieces of 4 x 4 x 2 mm CdZnTe (CZT) form a 1.2 x
0.6 m sensitive area in the detector plane. Groups of 128 detector elements
are assembled into 8 x 16 arrays, each connected to 128-channel readout Application
Specific Integrated Circuits (ASICs). Detector modules, each containing two
such arrays, are further grouped by eights into blocks. This hierarchical
structure, along with the forgiving nature of the coded aperture technique,
means that the BAT can tolerate the loss of individual pixels, individual
detector modules, and even whole blocks without losing the ability to detect
bursts and determine locations. The CZT array will have a nominal operating
temperature of 20 degrees C, and its thermal gradients (temporal and spatial)
will be kept to within +/- 1 degree C. The typical bias voltage is -200
V, with a maximum of -300V.
The BAT has a D-shaped coded aperture mask, made of ~54,000 lead tiles (5 x 5 x 1 mm) mounted on a 5 cm thick composite honeycomb panel, which is mounted by composite fiber struts 1 meter above the detector plane. Because the large FOV requires the aperture to be much larger than the detector plane and the detector plane is not uniform due to gaps between the detector modules, the BAT coded-aperture uses a completely random, 50% open-50% closed pattern, rather than the commonly used Uniformly Redundant Array pattern. The mask has an area of 2.7 m2, yielding a half-coded FOV of 100 degrees x 60 degrees, or 1.4 steradians. A graded-Z fringe shield, located both under the detector plane and surrounding the mask and detector plane, will reduce background from the isotropic cosmic diffuse flux and the anisotropic Earth albedo flux by ~95%. The shield is composed of layers of Pb, Ta, Sn, and Cu, which are thicker nearest the detector plane and thinner near the mask.
A figure-of-merit (FOM) algorithm resides within the BAT flight software and decides if a burst detected by the BAT is worth requesting a slew maneuver by the spacecraft. If the new burst has more "merit" than the pre-programmed observations, a slew request is sent to the spacecraft. Uploaded, rapid-reaction positions are processed exactly the same as events discovered by the BAT. The FOM is implemented entirely in software and can be changed either by adjusting the parameters of the current criteria or by adding new criteria.
BAT OperationsThe BAT runs in two modes: burst mode, which produces burst positions, and survey mode, which produces hard X-ray survey data. In the survey mode the instrument collects count-rate data in 5-minute time bins for 80 energy intervals. When a burst occurs it switches into a photon-by-photon mode with a ring-buffer to save pre-burst information.
Burst DetectionThe burst trigger algorithm looks for excesses in the detector count rate above expected background and constant sources. It is based on algorithms developed for the HETE-2 GRB observatory, upgraded based on HETE-2 experience. The algorithm continuously applies a large number of criteria that specify the pre-burst background intervals, the order of the extrapolation of the background rate, the duration of the burst emission test interval, the region of the detector plane illuminated, and the energy range. The BAT processor will continuously track hundreds of these criteria sets simultaneously. The table of criteria can be adjusted after launch. The burst trigger threshold is commandable, ranging from 4 to 11 sigma above background noise with a typical value of 8 sigma. A key feature of the BAT instrument for burst detection is its imaging capability. Following the burst trigger, the on-board software will check for and require that the trigger corresponds to a point source, thereby eliminating many sources of background such as magnetospheric particle events and flickering in bright galactic sources. Time-stamping of events within the BAT has a relative accuracy of 100 microsec and an absolute accuracy from the spacecraft clock of ~200 microsec. When a burst is detected, the sky location and intensity will be immediately sent to the ground and distributed to the community through the Gamma-Ray Burst Coordinates Network (GCN) (Barthelmy et al. 2000).
Above is an example of the response of the BAT to a simulated GRB. This shows a moderately difficult case: a GRB near the BATSE threshold (0.3
If there is no background subtraction, the resulting image will contain bright steady sources that can be confused with the transient. Images iii and iv (bottom middle and right) show that, in this simulation, the GRB is still detectable in the coarse and fine images, even though the steady sources are much brighter.
Hard X-ray SurveyWhile searching for bursts, the BAT will perform an all-sky hard X-ray survey and monitor for hard X-ray transients. The BAT accumulates detector plane maps every 5 minutes, which are included in the normal spacecraft telemetry stream. Sky images are searched to detect and position sources. The sensitivity of the survey is about 1 mCrab in the 15-150 keV band for 2 years. For regions where there are perpetually numerous strong sources in the BAT FOV (i.e. the Galactic Plane), the limiting sensitivity will be ~3 mCrab.
For on-board transient detection, 1-minute and 5-minute detector plane count-rate maps and ~30-minute long average maps are accumulated in 4 energy bandpasses. Sources found in these images are compared against an on-board catalog of sources. Those sources either not listed in the catalog or showing large variability are deemed transients. A subclass of long, smooth GRBs that are not detected by the burst trigger algorithm may be detected with this process. All hard X-ray transients will be distributed to the world community through the Internet, just like the bursts.
Detector PerformanceA typical spectrum of the 60 keV gamma-ray line from an 241Am radioactive source for an individual pixel is shown in the figure below. It has a full-width-half maximum (FWHM) at 60 keV of 3.3 keV (dE/E = 5%), which is typical of CZT detectors. Simulations have calculated an average BAT background event rate of 10,000 events s-1, with orbital variations of a factor of two around this value. This yields a GRB sensitivity of ~10-8 erg cm-2 s-1, 5 times better than BATSE.
The combination of the 4 mm square CZT pieces, plus the 5 mm square mask cells and the 1 m detector-to-mask separation gives an instrumental angular resolution of 20 arcmin FWHM, yielding a conservative 4 arcmin centroiding capability for bursts and steady-state sources given an 8 sigma burst threshold.
References"The Burst Alert Telescope (BAT) on the Swift MIDEX Mission"; Barthelmy, S. D. on behalf of the Swift Instrument Team, SPIE, vol 4140, pp 50, 2000.
"GRB Coordinates Network (GCN): A Status Report"; Barthelmy, S. D.; Cline, T. L.; Butterworth, P.; Kippen, R. M.; Briggs, M. S.; Connaughton, V.; Pendleton, G. N. Gamma-ray Bursts, 5th Huntsville Symposium, held in Huntsville, Alabama, USA, 18-22 October, 1999. (R. Marc Kippen, Robert S. Mallozzi, Gerald J. Fishman ed.) AIP Conference Series, Vol. 526. Published by American Institute of Physics, Melville, New York, 2000., p.731
If you have a question about Swift, please contact us via the Feedback form.